# Quiz Corner: Flowmeter accuracy statements

Oct. 17, 2017

What is the accuracy statement associated with this flowmeter?

#### A flowmeter is calibrated to its full scale of 200 flow units but is capable of up to 400 flow units. The flowmeter can measure a flow of 100 flow units within ±1 unit and can measure 200 flow units within ±2 units. What is the accuracy statement associated with this flowmeter?

A. ±1 percent of rate
B. ±1 percent of full scale
C. ±1 percent of calibrated span
D. ±1 percent of meter capacity

The full scale and calibrated span are both 200 flow units. The accuracy at 100 flow units expressed as a percentage of these parameters would be ±1/200 or 0.5 percent of full scale or calibrated span. Answers B and C are not correct.

At 100 flow units, the accuracy expressed as a percentage of meter capacity (400 flow units) would be ±1/400 or 0.25 percent of meter capacity. Answer D is not correct.

Similarly, the accuracy expressed as a percentage of the flow rates (100 and 200 flow units) is ±1/100 (and ±2/200) or 1 percent of rate, which confirms that Answer A is correct.

Note that "±1 percent" is not a complete specification. Be sure to ask, "Percentage of what?"

David W. Spitzer is a regular contributor to Flow Control magazine and a principal in Spitzer and Boyes LLC, which offers engineering, seminars, strategic, marketing consulting, distribution consulting and expert witness services for manufacturing and automation companies. Spitzer and Boyes is also the publisher of the Industrial Automation INSIDER. He has more than 40 years of experience and has written more than 10 books and 350 articles about flow measurement, instrumentation and process control.

Spitzer may be reached at 845-623- 1830 or via spitzerandboyes.com. Click on the "Products" tab to find his Consumer Guides to various flow and level measurement technologies.

### David W. Spitzer

David W Spitzer’s new book Global Warming (aka Climate Change): An Understandable Data-Driven Explanation and Pathway to Mitigation (Amazon.com) adds to his over 500 technical articles and 10 books on flow measurement, instrumentation, process control and variable speed drives. David offers consulting services and keynote speeches, writes/edits white papers, presents seminars, and provides expert witness services at Spitzer and Boyes LLC (spitzerandboyes.com or +1.845.623.1830).

### Micro Motion™ G-Series Compact, Drainable Coriolis Flow and Density Meters

Micro Motion G-Series Coriolis flow and density meters are ideally suited for Process Monitoring and Optimization applications, offering easy selection with pre-selected models...

### Learn about: Micro Motion G-Series Coriolis Flow and Density Meters

The Micro Motion G-Series is designed to help you access the benefits of Coriolis technology even when available space is limited.

### Micro Motion 4700 Coriolis Configurable Inputs and Outputs Transmitter

The Micro Motion 4700 Coriolis Transmitter offers a compact C1D1 (Zone 1) housing. Bluetooth and Smart Meter Verification are available.