# QUIZ CORNER: Volumetric Flowmeters

Feb. 21, 2008

The subject matter for flow measurement related to testing for a teaching position at a technical school was recently circulated. It included mass flowmeters (thermal and Coriolis) and volumetric

The subject matter for flow measurement related to testing for a teaching position at a technical school was recently circulated. It included mass flowmeters (thermal and Coriolis) and volumetric flowmeters (Annubar, nozzle, positive displacement, magnetic, turbine, vortex shedding, ultrasonic, Pitot, orifice plate, rotameter, Venturi, and V-cone). Which of the following groupings are correct for volumetric flowmeters?

A. Annubar, nozzle, Pitot, orifice plate, Venturi, V-cone
B. Magnetic, turbine, vortex shedding, ultrasonic
C. Positive displacement, magnetic, turbine, vortex shedding, ultrasonic, rotameter
D. Annubar, nozzle, positive displacement, magnetic turbine, vortex shedding, ultrasonic, Pitot, orifice plate, rotameter, Venturi, and V-cone
E. None of the above

Commentary
The equations for fluid flow in a pipe are:

Q = A * v
W = ρ * Q

where Q is the flowing volume, A is the cross-sectional area of the pipe, v is the velocity of the fluid in the pipe, W is the mass flowrate, and ρ is the fluid density.
Some flowmeters (volumetric) measure the flowing volume of the fluid while others measure the mass of the fluid. Other flowmeters measure the velocity of the fluid and infer the flowrate using the cross-sectional area of the pipe (A). Still other flowmeters infer the flow by measuring the velocity head (½ ρv2) across the flowmeter.

The flowmeters listed in Answer A are differential-pressure flowmeters. They present an obstruction to the flow that creates a differential pressure that is measured and used to infer the flow in the pipe. These flowmeters are not volumetric flowmeters, so Answer A is not correct.

The flowmeters listed in Answer B measure the velocity of the flowing fluid. They are not volumetric flowmeters. Thus, Answer B is not correct.

Regarding Answer C, positive-displacement flowmeters measure the volume of the fluid. However, the remaining flowmeters measure velocity. Therefore, Answer C is not correct. Similarly, Answer D is not correct. Thus, Answer E is correct.

Testing will apparently be administered based on this subject matter.

David W. Spitzer, P.E., is a regular contributor to Flow Control. He has more than 30 years of experience in specifying, building, installing, startup, and troubleshooting process control instrumentation. He has developed and taught seminars for over 20 years and is a member of ISA and belongs to the ASME MFC and ISO TC30 committees. Mr. Spitzer has written a number of books concerning the application and use of fluid handling technology, including the popular “Consumer Guide” series, which compares flowmeters by supplier. Mr. Spitzer is currently a principal in Spitzer and Boyes LLC, offering engineering, product development, marketing, and distribution consulting for manufacturing and automation companies. He can be reached at 845 623-1830.

www.spitzerandboyes.com

### Micro Motion™ G-Series Compact, Drainable Coriolis Flow and Density Meters

Micro Motion G-Series Coriolis flow and density meters are ideally suited for Process Monitoring and Optimization applications, offering easy selection with pre-selected models...

### Learn about: Micro Motion G-Series Coriolis Flow and Density Meters

The Micro Motion G-Series is designed to help you access the benefits of Coriolis technology even when available space is limited.

### Micro Motion 4700 Coriolis Configurable Inputs and Outputs Transmitter

The Micro Motion 4700 Coriolis Transmitter offers a compact C1D1 (Zone 1) housing. Bluetooth and Smart Meter Verification are available.